Dukus The fitting of curves ido specific sets of calibration or user data from flow measurement devices is covered in ISO and ISO The faster, easier way to work with standards. All standards are subject to revision, and parties to agreements based on this International Standard are encouraged to investigate the possibility of applying the most recent editions of the standards indicated below. BSI also holds the copyright, in the UK, of the publications of the internationalstandardization bodies. Click to learn more. Extra care should be taken during pipework construction to prevent excessive strain on the primary device, both during and after installation.
|Published (Last):||23 October 2009|
|PDF File Size:||19.72 Mb|
|ePub File Size:||11.71 Mb|
|Price:||Free* [*Free Regsitration Required]|
Mazulrajas These cables should be as short as possible and not exceed wn limit imposed by the manufacturer. This method is becoming less commonly used in modern electromagnetic flowmeters. General guidance on various aspects of the primary device is set out in 7. As the user knows the properties of the liquid to be metered, the ultimate decision on the materials to be used should rest with him. Remember me Forgot password? This International Standard ido flowmeter types in both a.
Some of the earliest applications were concerned with measuring the flow rate of bloodKolin Care should be taken to ensure that signal cables are not routed in proximity to high current cables. If the conductivity is not uniform throughout the meter, errors may also occur. This error comprises both systematic and random components depending upon the conditions of the flowmeter calibration, and can vary for each test point of the calibration curve. General introduction and classification. Your consent to our cookies if you continue to use this website.
Subscription pricing is determined by: As the voice of the U. Standards Subsctiption may be the perfect solution. Methods of evaluating the performance of electromagnetic flow-meters for liquids.
Flow pattern effects are described in 7. The meter samples the signal at zero magnetic field and zero adjusts, but does not differentiate against all bbs spurious signals. Means should be provided for adjusting and aligning the adjacent pipework. If necessary, warning devices should be installed to preserve measurement integrity.
To prevent the formation isi gas bubbles on the electrodes, these should normally be installed horizontally. The electromagnetic flowmeter consists of a primary device and one or more secondary devices. The shaft length and the frequency of ultrasound are chosen to produce an antinode at the electrode face. Subscribing members of BSI are kept up to date with standards developments and receive substantial discounts on the purchase price of standards. The same applies if the offset is in the negative direction.
Ensure use of the most current version of this document by searching British Standards Online at bsonline. In the case where cathodic protection is used to protect buried pipework, this precaution becomes essential see 7. The use of direct current as a means of exciting the magnet has been quickly discarded because of the difficulty of electrochemical and polarization emfs appearing at the electrodes, and for many years excitation by alternating current was the norm.
However, it was not until the s that industrial applications became a reality. The impedance of the liquid path should ideally be at least two orders of magnitude less than that of the input of the secondary device. Durchflussmessung von leitfaehigen Fluessigkeiten in geschlossenen Leitungen. We recommend that you check the website of the publishers of the international document before making a purchase. Up-to-date lists and bibliographical references concerning such national standards may be obtained on application to the Central Secretariat or to any CEN member.
When designing the piping system, access for installing and removing the primary device as well as access to the electrical connections should be provided. You may find similar items within these categories by selecting from the choices below:. One or more pairs of electrodes, diametrically-opposed, are set into the wall such that the diameter on which they lie, the magnetic field and the flow direction are mutually orthogonal.
Covers their installation, operation, performance and calibration. The coil assembly is either mounted externally or within the pipe. Instructions in relation to electrical grounding of the flowmeter system shall be rigidly observed. It is identical with ISO If, in this case, the mean axial velocity corresponding to the maximum flow-rate is less than that recommended by the manufacturer, a primary device with a smaller bore should be used.
Part 1BS If these details are to be used for any other purpose than implementation then the prior written permission ba BSI isl be obtained. The power supply should be from a point that is as free as possible from transient voltages. Systematic errors cannot be reduced by repeating measurements, since they arise from the characteristics of the measuring apparatus, the installation and the flow characteristics.
TOP Related Posts.
BS EN ISO 4064-5:2014
Yojas This approach is generally used on crystalline-type coatings. We have no amendments or corrections for this standard. The coils producing the magnetic field may be energized from the normal single-phase supply, or from some other supply. Its application is generally limited only by the requirement that the metered liquid shall be electrically conductive and non-magnetic. Bss the voice of the U. The faster, easier way to work with standards. A primary device with a bore smaller than that of the adjacent pipework may also be used for other reasons, e.
ILNAS EN ISO 6817
Ararr Equipotential conductive links usually copper braids should be fitted across both flange joints see Figure gs. Except as permitted under the Copyright, Designs and Patents Act no extract may be reproduced, stored in a retrieval system or transmitted in any form or by any means — electronic, photocopying, or otherwise — without prior written permission from BSI. Your basket is empty. Locations close to electrical equipment which may interfere with the flow measurement signal, or locations where currents may be induced in the primary device, should be avoided. If the field is perpendicular to an electrically-insulated pipe which contains the hs liquid and if the electrical conductivity of the liquid uso not too low, a voltage may be measured between two electrodes on the wall of the pipe. Extra care should be taken during pipework construction to prevent excessive strain on the primary device, both during and after installation.
BS EN ISO 6817:1997
Kagazshura It covers their installation, operation, performance and calibration. We recommend that you check the website of the publishers of the international document before making a purchase. Sio using electromagnetic flowmeters. It is essential that the primary device is correctly aligned on the pipe axis when it is bolted into the pipework. Please first verify your email before subscribing to alerts.